Flow Cytometry Filters
Figure 1. Diagram of a flow cytometer.
Image credit: Alluxa
Used across a variety of biological disciplines, fluorescence-based flow cytometers rapidly and accurately quantify cells and cellular components. As one of the most important components of these systems, flow cytometry filters must be specifically designed to maximize signal-to-noise ratios while minimizing crosstalk between fluorophores.
Flow cytometers are fluorescence-based systems that are used to quantify and sort cells that have been tagged with one or more fluorophores. Individual cells are first labeled with fluorescent tags, and then suspended in a physiological buffer, called sheath fluid, before they are injected into the fluidics system of the instrument. Inside the flow cell of the fluidics system, the suspended cells are arranged in a single-file row by means of hydrodynamic focusing. As the line of cells moves through the flow tube, each cell flows past one or more interrogation points where a laser beam is used to excite the fluorophores in the sample. The fluorescence emission signals from each fluorophore are then isolated by their respective emission filters and directed to the appropriate detector by means of a series of dichroic filters (Figure 1).
Flow cytometers use photomultiplier tube (PMT) detectors that convert light into electric current, which is then digitized for analysis purposes. As each cell flows past the interrogation point, an electric pulse is ultimately generated that corresponds to the signal intensity at each point in the cell's path over time as it passes the laser beam. Once the cell flows completely past the beam, the signal intensity returns to baseline until the next cell reaches the interrogation point. The pulses are then quantified in order to give an accurate count of cells or specific cellular components. In the case of cell sorters, cells are directed to appropriate vials based on which fluorophores are detected.
Flow Cytometry Filters
Flow cytometers use fluorescence emission filters and dichroic filters that are specifically designed to isolate the relatively weak emission signals of the target fluorophore for each channel, while blocking out the signals of neighboring fluorophores (Figure 2). Flow cytometry emission filters must have passbands with high transmission and low ripple that capture the emission peaks of each fluorophore. Emission filters must also be designed with very steep edges and relatively narrow bandwidths in order to block as much of the overlap between fluorophore emission as possible. Wide-range out-of-band blocking should extend beyond OD6 (-60dB or 0.0001% transmission), particularly at the excitation laser line.
Figure 2. Graph showing the ideal placement of flow cytometry filters used with four standard fluorophores. Fluorophore emission intensity is shown after excitation with a 488 nm laser.
Dichroic filters for flow cytometry must be designed with steep edges and minimal polarization splitting in order to efficiently direct the target fluorescence emission signals to the correct detector. Flow cytometry dichroic filters should also have a surface flatness specification that is suitable for use with lasers in order to minimize spot size distortion and beam deviation caused by reflected wavefront error (RWE).
Additionally, since the emission light will be transmitted through multiple filters prior to reaching the detector, all flow cytometry filters should be specified to produce minimal transmitted wavefront error (TWE) to prevent signal distortion. And all filters in these systems should have a high laser damage threshold (LDT).
Our flow cytometry filters are all hard-coated using an innovative SIRRUS plasma deposition process. This allows us to reliably and repeatedly produce high-performance laser clean-up, emission, and dichroic filters with steep edges, transmission that is typically greater than 95%, precision wavelength control, low TWE, and a high LDT. Our dichroic beamsplitters for flow cytometry are manufactured to achieve surface flatness that is 2λ P-V per inch or better, depending on the specific requirements of your flow cytometer.
Our knowledgeable engineers will work with you to design the optimal flow cytometry filters for your system based on your fluorophore selections, excitation lasers, and system configuration.
Alluxa designs and manufactures high-performance optical thin films that are used in a variety of applications. All of Alluxa's thin-film optical filters and mirrors are hard-coated using a proprietary plasma deposition process on equipment that was designed and built by our team. This allows us to reliably and repeatedly produce the same high-performance optical thin films over several different coating runs, which translates to consistent performance across all of your systems.
Contact Alluxa for more information at email@example.com
or visit our website at http://www.alluxa.com
All content copyright ©2018 Alluxa